All information products included in https://us-cert.cisa.gov/ics are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see https://us-cert.cisa.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 8.0
- ATTENTION: Low attack complexity
- Vendor: Becton, Dickinson and Company (BD)
- Equipment: Viper LT
- Vulnerability: Use of Hard-coded Credentials
2. RISK EVALUATION
Successful exploitation of this vulnerability could allow an attacker to access, modify, or delete sensitive information.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
The following versions of BD Viper LT, an automated molecular testing system, are affected:
- BD Viper LT system: All Versions 2.0 and later
3.2 VULNERABILITY OVERVIEW
The affected product is vulnerable due to the use of hard-coded credentials, which may allow a threat actor to access, modify, or delete sensitive information including electronic protected health information (ePHI), protected health information (PHI), and personally identifiable information (PII).
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: United States
BD voluntarily reported this vulnerability to CISA.
BD is working to remediate the hard-coded credentials vulnerability in the BD Viper LT system and is providing this information to increase awareness. The fix is expected in an upcoming BD Viper LT system Version 4.80 software release.
Additionally, BD recommends the following compensating controls for users of the BD Viper LT systems that utilize hard-coded credentials:
- Ensure physical access controls are in place and only authorized end-users have access to the BD Viper LT system.
- Disconnect the BD Viper LT system from network access, where applicable.
- If the BD Viper LT system must be connected to a network, ensure industry standard network security policies and procedures are followed.
For more information on this issue, please see the associated BD product security bulletin on the BD website.
CISA reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
CISA also provides a section for control systems security recommended practices on the ICS webpage on cisa.gov. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS webpage on cisa.gov in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to CISA for tracking and correlation against other incidents.
No known public exploits specifically target this vulnerability. This vulnerability is not exploitable remotely.
For any questions related to this report, please contact the CISA at:
Toll Free: 1-888-282-0870
CISA continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.
Please share your thoughts.
We recently updated our anonymous product survey; we'd welcome your feedback.