All information products included in https://us-cert.cisa.gov/ics are provided "as is" for informational purposes only. The Department of Homeland Security (DHS) does not provide any warranties of any kind regarding any information contained within. DHS does not endorse any commercial product or service, referenced in this product or otherwise. Further dissemination of this product is governed by the Traffic Light Protocol (TLP) marking in the header. For more information about TLP, see https://us-cert.cisa.gov/tlp/.
1. EXECUTIVE SUMMARY
- CVSS v3 7.5
- ATTENTION: Exploitable from an adjacent network/low attack complexity
- Vendor: OFFIS
- Equipment: DCMTK
- Vulnerabilities: Path Traversal, Relative Path Traversal, NULL Pointer Dereference
2. RISK EVALUATION
Successful exploitation of these vulnerabilities could allow an attacker to cause a denial-of-service condition, write malformed DICOM files into arbitrary directories, and gain remote code execution.
3. TECHNICAL DETAILS
3.1 AFFECTED PRODUCTS
The following versions of DCMTK, libraries and software that process DICOM image files, are affected:
- DCMTK: All versions prior to 3.6.7
3.2 VULNERABILITY OVERVIEW
The affected product’s service class provider (SCP) is vulnerable to path traversal, allowing an attacker to write DICOM files into arbitrary directories under controlled names. This could allow remote code execution.
The affected product’s service class user (SCU) is vulnerable to relative path traversal, allowing an attacker to write DICOM files into arbitrary directories under controlled names. This could allow remote code execution.
The affected product has a NULL pointer dereference vulnerability while processing DICOM files, which may result in a denial-of-service condition.
- CRITICAL INFRASTRUCTURE SECTORS: Healthcare and Public Health
- COUNTRIES/AREAS DEPLOYED: Worldwide
- COMPANY HEADQUARTERS LOCATION: Germany
Noam Moshe of Claroty reported these vulnerabilities to CISA.
CISA recommends users take defensive measures to minimize the risk of exploitation of this vulnerability. Specifically, users should:
- Minimize network exposure for all control system devices and/or systems, and ensure they are not accessible from the Internet.
- Locate control system networks and remote devices behind firewalls and isolate them from the business network.
- When remote access is required, use secure methods, such as Virtual Private Networks (VPNs), recognizing VPNs may have vulnerabilities and should be updated to the most current version available. Also recognize VPN is only as secure as its connected devices.
CISA reminds organizations to perform proper impact analysis and risk assessment prior to deploying defensive measures.
CISA also provides a section for control systems security recommended practices on the ICS webpage on cisa.gov. Several recommended practices are available for reading and download, including Improving Industrial Control Systems Cybersecurity with Defense-in-Depth Strategies.
Additional mitigation guidance and recommended practices are publicly available on the ICS webpage on cisa.gov in the Technical Information Paper, ICS-TIP-12-146-01B--Targeted Cyber Intrusion Detection and Mitigation Strategies.
Organizations observing any suspected malicious activity should follow their established internal procedures and report their findings to CISA for tracking and correlation against other incidents.
No known public exploits specifically target these vulnerabilities.
For any questions related to this report, please contact the CISA at:
Toll Free: 1-888-282-0870
CISA continuously strives to improve its products and services. You can help by choosing one of the links below to provide feedback about this product.
Please share your thoughts.
We recently updated our anonymous product survey; we'd welcome your feedback.